ZOOLOGY AND BOTANY, MICROSCOPY, ETC. 



763 



the same as Stand Aii, which only differs from Stand Ai (see October 

 Journal) in being a little smaller. The prongs of the foot are hinged, 

 and can be shut together, and the stage and mirror can be swung round, 

 for packing in box. The size of the box is '.) x 10 x 29 cm., and tbe 

 weight complete is 4*7 kilos. 



Fig. 162. 



Reichert's New Steinach Stand C.*— -This stand (fig. 163) is fitted 

 with a large circular brass stage of about 105 mm. diameter. The 

 projection of the inclinable upper part is considerable, and affords a 

 convenient grip in carriage. The tilt of the inclination is regulated by 

 a lever-clamp. The coarse-adjustment is by raek-and-pinion, and the 

 fine by micrometer screw. The Abbe condenser has a screw for quick 



* C. Reichert, Vienna, Catalogue, Mikroscope, No. 2G (190S) p. 22, fig. 9. 



:: K 2 



