CENTRAL SCIENTIFIC CO., CHICAGO, U. S. A. 



465 



No. 12802. 



Nos. 12810-11. 



STERILIZERS, ALL KINDS 



STERILIZERS. Autoclaves, see Autoclaves. 



12800. STEEILIZEE, Instrument, Electrically Heated, with three heats. Made of heavy copper, nickel- 

 plated and double tin lined with perforated tray. Provided with automatic cut-out which 

 cuts off the current if the sterilizer should run dry. The hinged lid is provided with a ven- 

 tilator which prevents dripping when opening the instrument. Operates on either A.C. or 

 D.C. Current consumption on low heat, 125 watts; on high heat, 500 watts. Size, 10x4%x2% 

 inches deep. Complete with five feet of attachment cord, connector, and lamp socket 



plug $24.00 



12802. STEETLIZEB, Instrument, Lewis Form, with lifting tray, for sterilizing small instruments in 

 pathological and bacteriological laboratories. The tray carrying the instruments is raised 

 clear of the hot water by opening the lid, and is held there until the lid is closed. Made 

 of copper nickel-plated. Dimensions, 10x5x3 inches deep; total height closed, 6 inches. Com- 

 plete with burner and base as illustrated 21.00 



STERILIZER, Combination Instrument and Dressing, with control valve, for sterilizing instru- 

 ments and dressings. Consists of a double walled chamber resting on a removable base 

 which is used to contain water for generating steam. The steam passes between the double 

 walls and after the contents of the inner chamber have become heated to 212F., the live 

 steam is admitted by a control valve at the top and forced through the contents and out at 

 the bottom of the other end. VThen sterilization is completed, the steam is again cut off and 

 the contents are left in the chamber until thoroughly dried. The base can be used separately 

 with the cover and one rack as an instrument sterilizer. Complete with two wire cloth racks 

 and valve opening for thermometer. 



No A 



Length, inches 14 



Width, inches 6 



Depth, inches 6 



Made of heavy tin, copper bottom 9.50 



Made of polished copper, tin lined 19.50 



STERILIZEK, Combination Instrument and Dressing, Electrically Heated, of polished copper, 

 tin lined, same as No. 12807, but with Prometheus electric heater mounted in the base. Can 

 be attached to any lamp socket. Complete with three-heat plugs, attachment cord and plug. 



No ...'. A B 



Length, inches 14 18 



12810. For 110 volts 38.00 



12811. For 220 volts 38.00 53.50 



12806. 

 12807. 



B 



18 



7% 



8 



12.00 

 26.00 



