CH. F] DRAWING WITH THE MICROSCOPE 



B 



I2 7 



FIG. 114. C 



Abbe Camera Lucida in position to avoid distortion. 



FIG. 114. The Abbe Camera Lucida with the mirror at 35. 



Axis, Axis. Axial ray from the microscope and from the drawing surface. 



A. B. Drawing surface raised toward the microscope 20. 



Foot. The foot or base of the microscope. 



Mirror with quadrant (Q). The mirror is seen to be at an angle of 35. 



Ocular. Ocular of the Microscope. 



P. P. Drawing pencil and the cubical prism over the ocular. 



W. Wedge to support the drawing board. 



A. Geometrical figure of the preceding, showing the angles made by the 

 axial ray with the mirror and the necessary elevation of the drawing board to 

 avoid distortion. From the equality of opposite angles, the angle of the axial ray 

 reflected at jj must make an angle of 110 with a horizontal drawing board. The 

 board must then be elevated toward the microscope 20 in order that the axial ray 

 may be perpendicular to it, and thus fulfill the requirements necessary to avoid 

 distortion ($ 779, 183}. 



B. Upper view of the prism of the camera lucid a. A considerable portion of 

 the face of the prism is covered, and the opening in the silvered surface appears 

 oval. 



C. Quadrant to be attached to the mirror of the Abbe Camera Lucida to in- 

 dicate the angle of the mirror. As the angle is nearly always 45 , 40, or 55, 

 only those angles are shown. 



