754 



A TEXT-BOOK OF BOTANY. 



contents, but which in focussing upon them are seen to be optical 



or microscopical sections of the pores. 



Micrometry or Microscopic Measurement. — In the micro- 

 scopic study of any substance a knowl- 

 edge of the comparative size of the 

 elements is often of much help in deter- 

 mining the identity of material under 

 examination, 'and for this reason the 

 student should early learn to measure the 

 characteristic elements, or those showing 

 a variation in size in different plants, as 

 starch grains, calcium oxalate crystals, 

 diameter of cells, thickness of cell-walls, 

 etc. The method best adapted for this 

 work is that involving the use of a micro- 

 metric scale which is placed in the eye- 

 piece and known as the ocular micrometer. 

 But to determine the value of the ocular 

 micrometer it is necessary to use another 

 scale known as the stage micrometer. 

 The stage micrometer, as its name indi- 

 cates, is used on the stage, and when 

 placed in juxtaposition to an object indi- 

 cates its size. However, it is obviously 

 impracticable always to place an object 

 alongside of the scale, and hence in prac- 

 tice the ocular micrometer is used, the 

 value of the divisions of which are 



scop?'showin|The°?e'iitk>n'^of determined by comparison with those of 



the ocular micrometer (o) and ^i . • , / t^' ^\ t-i 



the stage micrometer (s). As the stagc micromctcr (Fig. 417). The 

 oTthrScSlvllt\Qlve''e<^nWa. valuc of the divisious of the ocular scale 



lent to 4 divisions of the stage • r j*rc i. i, • i* 



micrometer, and thus each di- vanes for different objcctivcs, eye-pieces 



vision of the ocular is equiva- j i. u 1 i-t t. v • j. 



lent to 2 microns (see p. 813). and tubc lengths, hcucc it IS uccessary to 



d, diaphragm in eye-piece, i. • j.i 1 r i.u j* • • r xU 



on which the ocular microm- asccrtaiu the valuc of the divisions for the 

 different optical combinations and tube 

 lengths employed. The stage micrometer is usually divided into 

 tenths and hundredths of a millimeter, and the millimeter being 

 equivalent to looo microns (the micron being indicated by the 



