50 



Equipment of the Students' Laboratory and Students' Work Room. 



No. 50297 



50299. 1:31. 



50,297. Stink Cupboard with Cupboard Substructure, Figure, 1.15 m long, 0.7 m deep. s. d. 

 2.3 m high, framework of pitch-pine, with gas and water leads 12. 10. 



The gas and water leads are supplied ready mounted; the gas lead terminates in 2 hose unions 

 inside the cupboard, the taps being fitted outside on the front. The cupboard has a water tap 

 standard inside and a water outlet, and a water tap outside. 



The illustration differs somewhat in the construction of the leads. 



Stink Cupboards with a number of Compartments, Cupboard Substructure, Figure. 



List No. 



Length of Stink Cupboard 



Number of Compartments 



With Gas and Water Leads 



50,298 



o 



50,299 

 3 



2 



50,300 



3 

 3 



50,301 



4 

 3 



50,302 



4 m 

 4 



S 

 p 



o 



> 



(0 



3 



O 



$ 







and with Slate Table top / 



21. 10. 25. 10. 29. 15. 33. 12. 39. 15. 



The stink cupboards are supplied in. various lengths, 2.60 m high, 0.70 m inside depth, being 

 constructed of pine. They consist in the upper portion of 1 4 compartments, separated by glass 

 partition walls. The lower part contains double-doored cupboards, and has rounded base fillets of oak. 

 The table top is of oak, being let in with polished, acid-proof slate slabs. Each compartment has 2 gas 

 taps and 1 standard with water tap, lead water waste with strainer and lead waste pipe. All com- 

 partments have large sliding windows in front, each sliding window being provided with a small drop 

 window in order to allow of using the cupboard without the necessity of opening the large window. 

 Regarding other details of construction, see the introductory remarks on p. 48. 



It is advantageous to have a number of water outlets. 



If desired, one compartment of the cupboard can be fitted with a sand bath, No. 50,311 

 ( 1. 13. 0), and another with water bath, No. 50,312 ( 4. 4. 0). The sand bath consists of a flat box 

 of lead covered sheet iron, which is fitted in place of the slate slab, and which is heated under- 

 neath by gas. The water bath consists of a water box of metal, rings being let into the upper cover 

 plate. The box is heated by gas flames underneath. 



Iron Stink Cupboards (cf. Figs. 50,303 and 50,306 on p. 52). 



These cupboards combine ease of inspection with small 

 requisite. 



See introductory remarks on p. 48 for details of construction. 



amount of room 



Iron Stink 



Cupboards 



with one 



Compartment 



List No. 



Length m 



Depth m 



Height m 



Price with Leads 



Cl. 52CO. 



