Tables 124-125. 



TABLE 124. — Correction of the Barometer for Capillarity.* 



143 



Diameter 

 of tube 

 in mm. 



I. Metric Measure. 



Height of Meniscus in Millimeters. 



0.4 



0.6 



0.8 



1.0 



1.2 



1.4 



1.6 



Correction to be added in millimeters. 



1.8 



4 

 5 

 6 



7 

 8 



9 



10 



13 



0.83 



•47 

 .27 

 .18 



1.22 



0.65 

 .41 

 .28 

 .20 

 ■IS 



1-54 

 0.86 

 .^6 

 .40 

 .29 

 .21 



•IS 

 .10 

 .07 

 .04 



1.80 



1. 21 



0.82 

 .56 

 .40 

 .29 

 .21 

 •IS 



2. British Measure. 



Diameter 



of tube 

 in inches. 



Height of Meniscus in Inches. 



.01 



.02 



.03 



.04 



.05 



.06 



.07 



Correction to be added in inches. 



.08 



0.024 

 .oil 

 .006 

 .004 



0.047 

 .022 

 .012 

 .008 

 .005 

 .004 



0.009 



.019 

 .013 



.008 



.006 



.003 



.002 



.001 



0.092 



.045 

 .028 

 .018 

 .012 

 .008 

 .005 

 .004 

 .002 



0.116 



.059 



•037 

 .023 

 .015 

 .010 

 .007 

 .005 

 .003 



0.07S 

 .047 

 .029 

 .018 

 .012 

 .ooS 

 .006 

 .004 



059 

 03 s 

 022 

 014 

 010 

 C06 

 005 



0.042 

 .026 

 .016 

 .012 

 .007 

 .005 



* The first table is from Kohlrausch (Experimental Physics), and is based on the experiments of Mendelejefl and 

 Gutkowski fjour. de Phys. Chem. Geo. Petersburg, 1877, or Wied. Beib. 1877). The second table has been calcu- 

 lated from the same data by conversion into inches and graphic interpolation. 



Scheel und Heuse, Annalen der Physik, 33, p. 291, igio. 



Smithsonian Tables. 



