Depth, 

 m 



Differences, °C (STD/SV - Hydrocast) 



-4^30™" 



+8h3Qmin 



ghpniin 



1000 



-0.02 



0.00 



-0.01 



1200 



-0.03 



-0.02 



-0.02 



1500 



-0.03 



- 



0.01 



2000 



- 



- 



0.00 



2500 



- 



- 



0.01 



3000 



- 



- 



0.01 



3500 



- 



- 



0.01 



Note : The heading for each column of differences is the time in hours (h) 

 and minutes (min) that the STD/SV profile was taken with respect 

 to the applicable hydrocast. 



Table 7. Comparison of Gulf of Alaska STD/SV and hydrocast 

 temperature measurements. 





Depth, 





Temperature °C 









Reversing 





Number 



m 



STD/SV 



Thermometer 



Difference 



9 



1430 



2.20 



2.20 



0.00 



10 



3512 



1.44 



145 



-0.01 



40 



5042 



1.52 



1.53 



-0.01 







1.52 



1.57 



-0.05 



59 



4182 



1.46 



1.46 



0.00 



88 



3925 



1.44 



145 



-0.01 



90 



1005 



2.69 



2.69 



0.00 



Table 8. Comparison of Gulf of Alaska STD/SV and reversing 

 thermometer temperatures at maximum profile depth. 



22 



