TABLE 2 — CALIBRATION MATRIX 
-3.33 -0.099 0.058 -0.002 -0.090 -0.066 
-0.008 5.00 -0.002 0.084 -0.168 -0.196 
-0.071 0.048 -3.34 0.070 -0.014 0.058 
Calibration Matrix = (C; ,) = 
J 0.009 -0.033 0.103 4.99 -0.071 0.297 
-0.234 -0.367 0.291 -0.861 -9.99 -1.87 
0.079 -0.103 0.223 0.020 -0.034 -5.02 
where, 
Fy Fx Fy Fy F, are indicated forces in volts 
My, My, My My. M,, are indicated moments in volts 
y) 2 Fyn sake Boe Fyn FA are applied forces in pounds 
Mx, Mx a vd Mixa! Mya! Me, are applied moments in inch- pound 
2) Fan C; for j= 1, 3,5 are in volts per pound 
M_, Ma C;; for j = 2, 4,6 are in volts per inch-pound 
The arrays are arranged by the coupling of the force and moment pairs of the three flexures; 
i.e., Flexure 1 measures F, and My, Flexure 2 measures Fy and M,, and Flexure 3 measures 
F, and M,. 
165 
