D4 
able D2 
alibration Data for ee Inc. + SReeromasn ee Current ca 
4 eeny et ateccaL eee - - 
Serial Number Gain a Gain ne Gain Offset Lave cea ae Mee 
Number (m/s)/V (m/)/V % Diff. | Diff (m/s) | (m/s)/V 
$385 : 2321 0.823 0.009 0.907} 0.009} 10.207 -0.001 0.865} 1.084] Average 
2322 1.038 0.008 1.139] 0.008 Bre 730 Bea 001 1 a 0.008} Average 
$476 e 2331 1.125} -0.132] suspect Pre-cal. 
2332 1.253} -0.006} calibration Pre-cal. 
S760 X 2341 1.267) -0.013 1.290] -0.012 1. seal mat ail Pre-cal. 
Vf 2342 1.086} -0.015 1.095} -0.016 0.829 0.002 Pre-cal. 
S761 X 2351 1.583 0.032 switch| Pre-cal. 
Y 2352 1.581 0.012} to S762 Pre-cal. 
S762 X 2351 1.099] -0.051 1.122] -0.013 2.093 -0.041 Pre-cal. 
Y 2352 1.120 0.013 1.121} -0.004 0.089 0.019 Pre-cal. 
S765 X 1.154 0.023 1.177] -0.029 1.993 0.060 Pre-cal. 
Y 1.335 0.046 1.356] 0.067 1.573 -0.029) Pre-cal. 
S766 X 1.123 0.051 1.147) 0.051 2.137 -0.002) Pre-cal. 
Y 1.089 0.037 1.053] 0.127 -3.306 -0.094 Pre-cal. 
$837 . 0.941 0.029 1.230] 0.007} 30.712 0.018 Post-cal. 
0.949 0.025 1.0644 0.000] 12.118 0.024 Post-cal. 
$892 S 2311 1.277) -0.023 1.283] -0.015 0.470 -0.011 Pre-cal. 
2312 1.150 0.039 1.143] 0.040 -0.609 -0.001 Pre-cal. 
$972 X 2401 0.877] -0.299 1.064) -0.029] 21.323 -0.231 Pre-cal. 
Y 2402 1.048] -0.069 1.075] -0.032' 2.576 -0.038) 1.062 Average 
$1050 X 0.652] -0.014 0.650] -0.006 -0.307 -0.005 Pre-cal. 
Y 0.650] -0.014 0.649} -0.006 -0.154 -0.005 Pre-cal. 
$1080 y 1.093] -0.009) Post-cal. 
1.069} 0.006 Post-cal. 
$1013 x 2501 1.055} -0.001 1.0544 0.026 -0.095 -0.028 Pre-cal. 
2502 1.080 0.031 1.033} 0.030) -4.352 0.003 1.057} 0.031] Average 
$1015 _ 2801 1.036] -0.023} suspect Pre-cal. 
Y 2802 1.056} -0.004! calibration| Pre-cal. 
$1081 X 2301 1.000 0.002 1.010} 0.000 1.000 0.002 Pre-cal. 
Y 2302 1.029 0.001 1.039} 0.015 0.972 -0.015) Pre-cal. 
$1082 x 2201 1.013} -0.005 1.012] -0.003 -0.099 -0.002) Pre-cal. 
2202) 1.072} -0.193 1.065] -0.133 -0.653 -0.065 Pre-cal. 
$1083 x 2101 1.023} -0.013 1.018] -0.008 -0.489 -0.005) Pre-cal. 
2102 1.083} -0.012 1.053] -0.010) -2.770| -0.003) Pre-cal. 
$1084 X 2901 1.013] -0.011 Offset 
VV 2902 1.028] -0.052 1.040] -0.033 1.167 Pre-cal. 
$1012 X 2701 1.044 -0.029) 1.049] -0.038 0.479 -0.010 Offset 
Y 2702 1.089] -0.017 1.091] -0.021 0.184! -0.003) Pre-cal. 
Y 2602 1.051 0.052} destroyed Pre-cal. 
Appendix D Current Meter Calibrations 
