MAGNETIC INTENSITY. 503 
change which the cylinder had undergone in the preceding 
voyage, they persevered in diligently observing, and carefully 
recording, its time of vibration, at most of the principal ports which 
they visited in their voyage of five years’ duration. Nor was it 
until their return to the Cape Verd Islands, in September 1836, 
that they could infer, from observations repeated at the same spot 
as in their outward passage in 1832, that the cylinder had not 
varied in any thing like the degree that it had done in the prece- 
ding voyage, and that the care and pains they had bestowed were 
therefore likely to be recompensed by success. 
This appears a fitting opportunity to remark, how much the 
establishment in England of a depository for magnetic needles is 
needed ; whence officers, and persons desirous of making such 
observations, might be supplied with instruments, which had been 
kept a sufficient time to have attained their permanent magnetic 
state, and had been examined from time to time to prove that they 
had done so. The correction for temperature should be ascertained 
for each needle, and given with it; as well as the time of vibration 
(or whatever else constituted the measure of intensity,—as, for 
example, the angle of deflection in Mr. Lloyd’s statical needles, )— 
observed at the spot which should be selected as most suitable for 
a point of general comparison; and the observations should be 
repeated at the same spot on the return of the needle. The want of 
such an establishment has long been greatly felt ; and opportunities, 
where nothing was wanting but proper instruments, have been lost 
in consequence, where determinations of great value might have 
been obtained, in parts of the world of the highest magnetic interest, 
and where such opportunities are of rare occurrence. 
The corrections necessary to render the times of vibration at the 
different stations strictly comparable with each other, are as 
follows. 
Ist, For the rate of the chronometer. 
2d, For the temperature of the needle. 
3d, For the arc of vibration. 
Ath, For any change in the magnetic condition of the cylinder. 
In extensive voyages, the last-named correction, or that for the 
change in the cylinder itself, is the one which requires principal 
consideration. The corrections for temperature, and for the arc, 
on the first of which particularly much stress has sometimes been 
