Comparisons of Mercury and Platinum Thermometers. 37 

 Table XIII. 



Date No - of 



! Observations 



1 



Temp, on 



Callendar- 



Griffiths 



Air-Scale. 



Temp, on 



Rowland's 

 Air-Scale. 



Difference 

 A. 



Comparison of Platinum Thermometer with Baudin 6163. 



May 18th. 



5 



5-261 



5-249 



+0-012 





6 



9937 



9-922 



+0015 









15-340 



15-325 



+0-015 





6 



20-066 



20052 



+0-014 





4 



24-881 



24-861 



+0020 





5 



34916 



34-921 



-0-005 





4 



40-051 



40052 



-0001 



Comparison of Platinum Thermometer with Baudin 6163. 



May 24th. 



6 



9-997 



y-998 -0-001 





6 



20055 



20-056 -0-001 





6 



25-228 



25-222 



+0006 





6 



34-899 



34-909 



-0010 





6 



40283 



40-305 



-0-022 



Comparison of Platinum Thermometer with Baudin 6166. 



Apr. 10th. 



5 



10-873 



10844 



+0029 





5 



19-650 19-611 



+0-039 





5 



30136 



30114 



+0-022 



Comparison of Platinum Thermometer with Baudin 6166. 



May 21st. 



5 



5-131 



5127 



+0004 





6 



9-869 



9-852 



+0-017 





6 



14-899 



14-871 



+0-028 





7 



19-951 



19-923 



+0-028 





6 



24-987 



24-965 



+0022 





6 



30-226 



30-212 



+0-015 



Comparison of Platinum Thermometer with Baudin 6166. 



May 28th. 



6 5-242 



5-227 



+0015 





6 10091 



10066 



+0025 





6 



19-978 



19-937 



+0-041 





4 



30-191 



30-163 



+0-028 



Comparison of Platinum Thermometer with Baudin 6165. 



Apr. 14th. 



5 10-129 



10104 



+0-025 





5 20-274 



20-247 



+0027 





5 29735 



29-709 



+0-02I 1 . 



