Page A I - 4 
Design Handbook on Animal Research Facilities Using Static Microisolators 
Thermocouple Reading, F vs. Predicted Temperature, F 
5 
& oZ 
u 
3 oU 
If® no _ 
fea /O 
<y 
1 76 
H A - 
/ 4 
H 11 
ro 72 
Si i n 
y /O 
(if 
0 - 06 - 
6 
0 6 
5 7 
Then 
H Therm: 
0 7 
nocouple Readii 
meter, F O P red Tern, 
5 8 
igs,F 
F 
0 8 
2.1.2 Air Velocity Sensor Calibration 
Every morning turn on the sink in the back of the fan test chamber. Allow the water to run for at 
least 10 minutes. This is necessary because the cooling unit will overheat if hot water is 
introduced into it. 
Remove the large duct on top of the blue chamber. Open the damper below it. Replace the duct. 
Check the barometric pressure and the zero of the pressure gauge. 
Turn on the long gray chamber using the computer 
Pick “2” (Controls) [Return] 
Pick “1” (Run/Standby) [Return] 
Pick “1” (Run) [Return] [Return] 
Allow the blue chamber to cool for at least 10 minutes. 
Remove the large duct again and open the blue foam damper on it. 
Close the damper, which was below it. 
Turn on the calibrator fan and adjust it to the desired pressure (listed pressure + zero pressure). 
