Appendix I 
Page A I- 31 
3.2.2 Calorimeter Approach Air Velocity Calibration: ( Calibration Data Prior to Test 
2 ) 
Calorimeter 
n 
69 
48 
50 
47 
56 
Av. = 
Air velocity = 
55 
44 
47 
51 
51 
(fpm) 
51 fpm 
64 
48 
47 
45 
46 
Voltage Regulated Setting = 106 
Calorimeter 
n 
67 
52 
53 
54 
65 
Av. = 
Air velocity = 
65 
43 
40 
58 
62 
(fpm) 
54 fpm 
71 
43 
40 
54 
47 
Voltage Regulated Setting = 62 
Calorimeter 
ff3 
63 
45 
40 
49 
64 
Av. = 
Air velocity = 
51 
44 
39 
51 
59 
(fpm) 
50 fpm 
60 
41 
39 
45 
55 
Voltage Regulated Setting = 61 
3.2.3 Calibration of Humidity Sensors: ( Calibration Data Prior to Test 1) 
120% 
i 
Calorimeter#! 
/ 
A 
r Ou /C 
2 * 
“ AH®, 
A 
O 2 
3 § 40% 
"Sd 5 
® 20% 
0% 
i 
i 
y 
= 0.21 
51x - 0 
.1012 
R 2 =0.9986 
1 1 
3 
Sens 
l 3 4 5 
ior output, V 
6 
