Appendix I 
Page A I- 37 
3.2.7 Calorimeter C02and O 2 Recovery Calibration: ( Calibration Data Prior to Test 
1 ) 
Cal 
orimeter # 
Time (min) 
co 2 
(percent) 
Room C0 2 
(percent) 
Gen. C0 2 a 
(L/min) 
o 2 
(percent) 
Room 0 2 
(percent) 
Con 0 2 b 
(L/min) 
Ethanol wt c 
(g) 
0 
1.00 
0.10 
0.0497 
19.2 
20.5 
0.072 
156.64 
10 
1.00 
0.10 
0.0497 
19.2 
20.5 
0.072 
156.17 
20 
1.00 
0.10 
0.0497 
19.2 
20.5 
0.072 
155.67 
30 
1.00 
0.10 
0.0497 
19.2 
20.5 
0.072 
155.20 
40 
1.00 
0.10 
0.0497 
19.2 
20.5 
0.072 
154.72 
50 
1.01 
0.10 
0.0502 
19.1 
20.5 
0.077 
154.27 
60 
1.01 
0.10 
0.0502 
19.1 
20.5 
0.077 
153.72 
Flow rate=5.52 1/min 
Calorimeter #2 
Time (min) 
co 2 
(percent) 
Room C0 2 
(percent) 
Gen. C0 2 a 
(L/min) 
o 2 
(percent) 
Room 0 2 
(percent) 
Con 0 2 b 
(L/min) 
Ethanol wt c 
(g) 
0 
1.13 
0.11 
0.0563 
19.0 
20.4 
0.0773 
167.35 
10 
1.13 
0.11 
0.0563 
19.0 
20.5 
0.0828 
166.84 
20 
1.14 
0.11 
0.0569 
19.0 
20.5 
0.0828 
166.32 
30 
1.14 
0.11 
0.0569 
19.0 
20.5 
0.0828 
165.82 
40 
1.13 
0.11 
0.0563 
19.0 
20.5 
0.0828 
165.32 
50 
1.13 
0.11 
0.0563 
19.1 
20.5 
0.0773 
164.82 
60 
1.14 
0.11 
0.0569 
19.0 
20.6 
0.0883 
164.32 
70 
1.14 
0.11 
0.0569 
19.1 
20.6 
0.0828 
163.81 
Flow rate = 5.52 
/min 
Calorimeter #3 
Time (min) 
co 2 
(percent) 
Room C0 2 
(percent) 
Gen. C0 2 a 
(L/min) 
o 2 
(percent) 
Room 0 2 
(percent) 
Con 0 2 b 
(L/min) 
Ethanol wt c 
(g) 
0 
1.28 
0.07 
0.0668 
18.9 
20.6 
0.0938 
167.65 
10 
1.29 
0.07 
0.0673 
18.9 
20.6 
0.0938 
166.97 
20 
1.29 
0.07 
0.0673 
18.9 
20.6 
0.0938 
166.30 
30 
1.29 
0.07 
0.0673 
18.9 
20.6 
0.0938 
165.62 
40 
1.29 
0.07 
0.0673 
18.9 
20.5 
0.0883 
164.95 
50 
1.29 
0.07 
0.0673 
18.9 
20.5 
0.0883 
164.29 
60 
1.29 
0.07 
0.0673 
18.9 
20.5 
0.0883 
163.58 
70 
1.29 
0.07 
0.0673 
19.0 
20.5 
0.0828 
162.90 
Flow rate=5.52 1/min 
