Appendix I 
Page A I- 39 
3.2.8 Calorimeter CO 2 and O 2 Recovery Calibration : (Calibration Data Prior to Test 
2 ) 
Ca 
orimeter # 
Time (min) 
co 2 
(percent) 
Room C0 2 
(percent) 
Gen. CO/ 
(L/min) 
0 2 
(percent) 
Room 0 2 
(percent) 
Con 0 2 b 
(L/min) 
Ethanol wt c 
(g) 
0 
1.53 
0.06 
0.0697 
18.8 
20.7 
0.0901 
153.83 
10 
1.53 
0.06 
0.0697 
18.6 
20.6 
0.0949 
153.15 
20 
1.53 
0.06 
0.0697 
18.6 
20.6 
0.0949 
152.49 
30 
1.53 
0.06 
0.0697 
18.6 
20.7 
0.0996 
151.83 
40 
1.53 
0.05 
0.0702 
18.6 
20.7 
0.0996 
151.16 
50 
1.54 
0.05 
0.0707 
18.5 
20.6 
0.0996 
150.49 
60 
1.54 
0.05 
0.0707 
18.6 
20.6 
0.0949 
149.82 
70 
1.54 
0.05 
0.0707 
18.6 
20.7 
0.0996 
149.15 
Flow rate = 4.743 1/min 
Calorimeter #2 
Time (min) 
C0 2 
(percent) 
Room C0 2 
(percent) 
Gen. C0 2 a 
(L/min) 
0 2 
(percent) 
Room 0 2 
(percent) 
Con 0 2 b 
(L/min) 
Ethanol wt c 
(g) 
0 
1.53 
0.05 
0.0740 
18.6 
20.7 
0.1050 
163.95 
10 
1.53 
0.05 
0.0740 
18.6 
20.7 
0.1050 
163.26 
20 
1.53 
0.05 
0.0740 
18.6 
20.7 
0.1050 
162.58 
30 
1.53 
0.05 
0.0740 
18.6 
20.8 
0.1100 
161.89 
40 
1.54 
0.05 
0.0745 
18.6 
20.8 
0.1100 
161.18 
50 
1.54 
0.05 
0.0745 
18.6 
20.8 
0.1100 
160.51 
60 
1.54 
0.05 
0.0745 
18.6 
20.8 
0.1100 
159.81 
70 
1.54 
0.05 
0.0745 
18.6 
20.8 
0.1 100 
159.12 
Flow rate = 5 1/min 
Calorimeter #3 
Time (min) 
co 2 
(percent) 
Room C0 2 
(percent) 
Gen. CO/ 
(L/min) 
o 2 
(percent) 
Room 0 2 
(percent) 
Con O/ 
(L/min) 
Ethanol wt c 
(g) 
0 
1.57 
0.05 
0.0608 
18.7 
20.6 
0.0760 
151.03 
10 
1.57 
0.04 
0.0612 
18.6 
20.6 
0.0800 
150.41 
20 
1.57 
0.04 
0.0612 
18.7 
20.6 
0.0760 
149.82 
30 
1.57 
0.04 
0.0612 
18.6 
20.7 
0.0840 
149.21 
40 
1.57 
0.04 
0.0612 
18.7 
20.6 
0.0760 
148.62 
50 
1.58 
0.03 
0.0620 
18.6 
20.8 
0.0880 
148.03 
60 
1.58 
0.03 
0.0620 
18.6 
20.8 
0.0880 
147.43 
70 
1.58 
0.03 
0.0620 
18.6 
20.8 
0.0880 
146.83 
Flow rate = 4 1/min 
