THE CONSTANTS OF THE CUP AN EMOMETER. 
803 
B=6 oz. ; /+B'= 1349-9 ; June 3, Bar 29'56 ; Therm. 6U5. 
No. 
D. 
S. 
N. 
A. 
b. 
V. 
V. 
W. 
/'"• 
Log. p. 
m. 
175 
22 
240 
83-5 
66"3 
118 
13-53 
1-18 
1-28 
351 
0 01060 
10-245 
176 
28 
102-2 
165 
91-2 
16-56 
2-945 
1-66 
56-25 
5 060 
177 
36 
124 
279 
84-8 
20-09 
4-98 
1-78 
88-8 
99 
3-676 
178 
46 
146 
393-6 
65-2 
23-66 
7-03 
2 39 
134-7 
3 028 
179 
56 
164 
481-4 
66 
26-57 
8 59 
2-29 
180 
2 826 
180 
66 
183 
570-7 
59-4 
2965 
10-20 
2 55 
245-4 
2-6545 
181 
75 
198-5 
631-3 
58-2 
32-16 
11-27 
2 60 
304-1 
” 
2623 
B= 
= 9 oz. 
;/+ B 
'= 18847. 
No. 
D. 
S. 
N. 
A. 
b. 
V. 
V. 
W. 
f". 
Log. p. 
m. 
182 
28 
240 
100-5 
45 
99-2 
16-28 
0-82 
1-52 
54-7 
0 01060 
18376 
183 
36 
123-5 
168-5 
90-2 
20-01 
3-01 
1-63 
88 
6 096 
184 
47 
148 
337-5 
72 
23-98 
6-02 
2-10 
136-6 
3633 
185 
57 
167 
436 
66-4 
27-06 
7-78 
2-27 
189-3 
3-185 
186 
66 
183 
517-7 
62-8 
29-65 
924 
2-41 
245-4 
2-948 
187 
75 
99 
197 
585-2 
56-2 
31-92 
10445 
2-69 
297-5 
If 
2-798 
B=12 oz. ; /+ B' = 2397-5. 
No. 
D. 
S. 
N. 
A. 
b. 
Y. 
V. 
W. 
/"'• 
Log. p. 
771 . 
188 
36 
240 
126 
142-3 
94-4 
20-42 
2-54 
1-65 
92 4 
0-01060 
7-397 
189 
47 
150-3 
310-4 
74 
24-35 
5-54 
2-04 
1421 
4-0265 
190 
57 
166‘5 
369 
69-4 
26-98 
6-59 
2-18 
187-8 
3 765 
191 
66 
177 
431 
63 
28-68 
7-69 
2-40 
225'5 
3-416 
192 
76 
99 
198 
530 
57 
32-08 
9-46 
2-65 
302 1 
99 
3111 
B=15 oz. ; /+B , = 2896:5. 
No. 
D. 
S. 
N. 
A. 
b. 
V. 
V. 
W. 
yv// 
Log. p. 
m. 
193 
47 
240 
146 
195 
76-4 
23-66 
3-48 
1-98 
131-5 
0 01060 
6-228 
194 
57 
164 
332-9 
68 2 
26-57 
5-94 
2-22 
180 
4-100 
195 
66 
99 
179 
408 
61-8 
29-00 
7-28 
2-45 
232 
99 
3-646 
196 
76 
19 
196 
496 
60-8 
31-76 
8-85 
2 49 
294 
” 
3306 
B= 18 oz. ; /■+ B' = 3390-5. 
No. 
D. 
S. 
N. 
A. 
b. 
y. 
V. 
W. 
/"'■ 
Log. p. 
m. 
197 
57 
240 
164-4 
274-3 
67”6 
26-64 
1-90 
2-24 
1813 
0 01060 
4-984 
198 
66 
99 
181-3 
384 
62-2 
29-375 
6-85 
2-43 
239 5 
3 931 
199 
76 
180 
147 
339-5 
56-6 
31-76 
8-08 
2-67 
293 
3599 
The results with No. IY. are less reliable than any of the others; the shorter arms 
and smaller cups having less power to overcome the casual irregularities of friction. 
