Almindeligt Drikkevand foraarsager følgelig ingen merkelig 
Fejl, og heller ikke smaa Fejl i Burettens Inddeling. 
I det Tilfælde, at Regnhøjden, ved Fordunstnings- 
maaleren Å, er større end Fordunstningshøjden, og der 
saaledes maa tages noget Vand ud af Fordunstningsskaalen 
for at faa Instrumentet paa Merke, er dette Vand salt og 
har en højere specifisk Vægt.  Sættes denne, med en tem- 
melig høj Værdi, til 1.027, den med Pipetten optagne 
Vandmængde til 97 Kubikcentimeter (som Tiltældet var 
den 9de Juni 1876) faaes: 
Q—e = 9 (1.027—1) MM »< O02 2.62: 
2.62 Gram eller Kubikcentimeter Vand svarer til en For- 
ae eller 0.7"12. 
ner Nojagtighedens Grænse. 
dunstningshøjde af Denne Correction er 
3. Virkningen af Temperaturens Forandring 7 det 
ydre Vand, hvort Bøjen flyder. Kaldes Vægten af Bøjen 
med Stet, Skaal og Vand, naar Instrumentet staar paa 
Merke: p, Bojens Rumfang under Merket, ved 0° C: Vi, 
den specifiske Vægt af Vandet, i hvilket Bøjen flyder, 
ved 09: S, dets specifiske Vægt ved f°: S,, Udvidelsescoeffi- 
cienten for 1° C af Bøjens Materiale: «, og Forskjellen i 
Apparatets Vægt i Vand af 0° og i Vand af 19, Merket i 
saa har man 
Vandkanten: 7, 
Vo S = p. 
Vo (Å + dat) 8, = p—x 
Ved Subtraction: Vy (59 -- (I + 3at) 8) = x 
og ne Vy Sy LE Es (1 + 3a DI 
0 
Forholdet S;: So findes af H. Tornøes Tabel i hans 
Afhandling ,Om Saltholdigheden af Vandet 1 det norske 
Nordhav* Side 52. Kaldes Volumet ved 0": v,. Volumet 
ved 7: v;, saa har man 
Io db Sad 
Ve Vi 
Ved Fordunstningsmaaleren Å er Vy 6560 Kubikcen- 
timeter, og med Sy = 1.027, faar man Vo Sy = 6735 Gram. 
Ved 8 ce % = 4D dom o8 KM S 205 & IDE 
Jernets lineare Udvidelsescoefficient er 0.0000122, bliver 
3 a@ = 0.0000366. Med disse Verdier faaes: 
For A, 2, = 6735 | — 2 (1 + 0.0000366 DÅ 
VE 
For B, « = 2785 (! — ~ (1 + 0.0000366 D) 
Den følgende Tabel giver Værdien af x for hver 
Grads Temperatur mellem 0° og 209 UC. 
water causes no 
in the graduation 
of 0.9982. Hence, common drinking 
appreciable error, nor do trifling errors 
of the burette. 
When the depth of fallen rain in the atmometer A 
exceeds that of the water evaporated, and water has to be 
taken from the evaporating dish to restore the equilibrium, 
this water is salt and has a higher specific gravity. Put- 
ting this specific gravity at 1.027 (a rather high value), 
the quantity of water taken up with the pipette at 97 
cubic centimetres (as actually observed June 9th 1876), 
we have 
Q—¢ = 97 (1.027—1) = 97 x 0.027 = 2 62. 
Now, 2.62 grammes or cubic centimetres of water correspond 
2.62 , 
9915” OF (PAN. 
approximates the limit of accuracy. 
bo 
to a depth of evaporation of This correction 
3. The effect of change of temperature of the outer 
water im which the buoy is floating. Let p be the weight 
of the buoy, along with the stem, dish, and water, when 
the instrument is adjusted to the mark, V, the volume of 
the buoy below the mark at 0" C, S the specific gravity, 
of the water in which the buoy floats at 09 UO, 8S, its 
specific gravity at f° C, aæ the coefficient of dilatation for 
1° C of the material of which the buoy is made, and ~, 
the difference in the weight of the apparatus in water of 
09 and in water of £ (the mark on the stem at the surface), 
we have 
Yo Sy =p 
Yi+3sat)S=p—-x, 
By subtraction V, (So —(1+ 3at) Si) Se 
Si 
g + 3«at)|. 
ang g,= VS, [1 — 
The ratio S,: S, is found from H. Tornøe's Tables 
in his Memoir on “The Amount of Salt in the water of 
the Nowegian Sea”, p. 52. Calling the volume at 0°, 2%, 
the volume at 7", v,, we have 
Qi 
S; 
Vi Sa 
Ss Of 
Vo 
and a, = V, 8, 1-1 + 3at)|: 
Ay 
In the atmometer A, V, is 6560 cubic centimetres, 
and with S,= 1.027; we have V, S,=6"35 grammes. In 
the atmometer B, V,= 2712 ccm. and V,S,=2785 gram- 
mes. The coefficient of linear dilatation bemg for iron 
0.0000122, 3 «= 0.0000366. These values give 
For atmometer A, 7, = 6735 
I — (1 +4 0.000036 DI 
t 
For B, a, = 2785 | 1 — "(1 4-0.0000366 0) 
t 
The following Table gives the value of x, for every 
degree of temperature between 09 and 20° C. 
